FarmaDolor : How to use the FarmaDolor
Your Farmadolor is at the cutting-edge technology, it respects the existing standards in special medical application.
The user interface has been made as simple as possible in limiting itself to two buttons: one to start, the other to adjust the power.
1. START “”
One short press on the key “” and the device turns on. A second press on the key “” and the device turns off.
2. SELECTING THE POWER “P”
The selection of the power level is indicated by the switching of the indicator light located above the key “P”.
By pressing on the key “P”, the indicator light turns off accordingly:
- No press: minimum intensity (the indicator light n°1 turns on)
- 1 press: medium intensity (The indicator light n°2 turns on)
- 2 successively press: maximum intensity (The indicator light n°3 turns on)
- 3 successively press: back to minimum intensity (The indicator light n°1 turns on)
Do not use the power 3 (max.) on the neck, calves, shoulders, forearms.
Note: pause for a few seconds the camera takes place during the program each frequency change.
3. END OF THE PROGRAM / STOPPING
- The device automatically stops after approximately 60 minutes.
- Emergency stop: press the start/stop key “”.
Your device has electrodes disconnection detector with the body. It stops by itself after 4 minutes with no use.