Size (unisex) From 34 to 60
Length of the unfolded belt 120 cm
Fabric 94 % polyester, 6 % elastane
2.Mains power supply
AC/DC 100-240V - 50/60Hz - 0.25A 5V 600mA
Only use the power supply provided with the device.
Please replace it in the event of damage.
Class I medical device (Directive 93/42/CEE)
Considered as BF applied parts.
Adhesive, hydrophilic and hypoallergenic electrodes.
Contact area 67.5 x 47 mm (x 2)
Estimated life expectancy 40 uses
Please order new GlObalStim-compatible adhesive electrodes on www.sport-elec.com
Class IIa medical device in accordance with Directive 93/42/CEE.
Software version 1
Type of current Symmetrical biphasic rectangular
Number of programs 8 programs
Number of adjustable independent channels 2
Current from 0 to 60 mA - from 0 to 60 Vcc - from 0 to 1000 Ω
Frequency range/Pulse width range From 1 to 100 Hz / De 160 à 320 μs
Power Rechargeable LiPo battery. 3.7V - 420mAh
Unit dimensions and weight 78 x 52 x 20 mm - 51 g
Automatic stop at the end of each program Yes - after approx. 4 min 30 sec of inactivity
Activation safety 100 %
Detection absence of contact Yes - independent detection on each of the 2 outputs
Indicator of power LCD display: 2 graduated scales
Indicator of low and/or defective battery Yes - LCD display
Indicator of programs LCD display: number 1 - 2 - 3 - 4 - 5 - 6 - 7 - 8
Temperature/Relative humidity for storage 0 °C to + 40 °C / 10 % to 90 %
Temperature/Relative humidity for use + 5 °C to + 40 °C / 20 % to 65 %
Connection of the electrode 2 snap fasteners per electrode
Warning Read the instructions carefully.
Conformity CE 0459 - 92/42/EEC
Guarantee/Life cycle 2 years
Appareil a but non thérapeutique
|CE marking concerning medical appliances.||Device in contact with the patient,
excluding the heart area.
|These units are free of materials likely to have a negative impact on the environment||Caution :
Read the instructions for use.
|Maximum intensity per channel
up to 1 kΩ : 60 mA.
|Relative humidity for storage :
from 10 to 90%.
|Temperature for storage :
from 0 to 40° C.
|Name of the manufacturer bringing the device
to the market.
|Please contribute to protect the environment
by observing the disposal instructions.
|The manufacturer respects the environmental law and contributes to the cost of recovery and recycling.||Follow the user manual instructions.||
Do not use if you carry a pacemaker.
Do not use if you are pregnant.
|This device is intended for domestic use.|