Body Beautiful : Starting the unit
Presentation of the unit
Your electromusculation device is an hight-tech appliance, it meets the current requirements in the field of medical application.
- The interface « man machine » was simplified to the maximum; it is realized from lightened pictograms, indicators and function key.
- Among the eight pictograms the one enlightened in green displays you the program chosen by the « P » key.
- Two scales indicator (1 per channel) displays you 72 levels of power.
- You will find also 4 control keys (see diagram page 3) :
- 1 « » key red - press shortly
- 1 « P » key to choose the programs - press shortly
- 2 double keys « +/- » to adjust the power of stimulation (press shortly).
A long press (1 second) on the key « - » returns the chosen power rapidly to 0.
Using the unit
Step 1 - ON / OFF «»
Short press on the « » key puts the device under or except power.
Every time when the power is on, the device carries out an auto-test on 11 essential safety parameters.
During the auto-test all the indicators are lit during ½ second.
If an internal defect in the device is detected, all the pictograms flash in green and the device stops. Then, return the device to After Sales Service.
The auto-test does not take into account the detection of the used batteries. When batteries are used, the auto-test is not activated.
A long press (of more than 3 seconds) involves the release of the auto-test. In this case stop the device, please, by pressing the « » key and then make a short press on the « » key.
Step 2 - Program selection key « P »
To select a program, make one or more short presses on the « P » key, the programs scroll.
The programs are announced by pictograms.
As soon as a program is selected, corresponding pictogram lights.
Once the program is chosen, you can start it by increasing the power on one of power keys « ».
During the program the « P » key is inactive.
Step 3 - End of the program - Stop during the program
At the end of the session the device stops automatically.
To stop the device during the program, put the 2 power keys back on zero or press the « » key.
Step 4 - Adjustment of the power
The power is divided into 10 levels : 5 green, 3 orange and 2 red levels.
To change the level press the « » key. Press 1 for both the bearing from 0 to 1, then 8 consecutive presses to the following power levels.
You have 72 levels to regulate the power from 0 to 72.
Step 5 - Batteries
If the batteries are good, there is no blinking.
If the batteries are weak, two orange indicators blink. Your device still functions.
If the batteries are used, two red indicators blink. Your device stops, even in the course of a program. Change the batteries.
Step 6 - Disconnection
Your device is equipped with a detector of electrodes disconnection.
If the contact between the human body and a single electrode is absent, the power is stopped on the concerned channel. You are warned of a disconnection by blinking of green indicator during 30 seconds.
If none of the electrodes is in contact with the body, the power cannot be emitted.